The basic Wheatstone bridge circuit consists of two arms composed of two resistors in series. A source of excitation (voltage or current) is connected and the output is the difference between the outputs of the two voltage dividers. The Wheatstone bridge circuit is useful for measuring small changes in resistance like that of the strain gauge where typically less than 1% change occurs.

The voltage output of the wheatstone bridge V_{G} is given by

If the bridge is balanced V_{G} = 0 since all the resistances are equal. When a change is applied to R_{X} , R_{X} becomes R + ΔR and substituting R to all the other resistances, the change in output voltage ΔV_{G} is

With advertising revenues falling despite increasing numbers of visitors, we need your help to maintain and improve this site, which takes time, money and hard work. Thanks to the generosity of our visitors who gave earlier, you are able to use this site at no charge.

If you have benefited from this site and are able, please
**give $10 via Paypal**. It will allow us
continue into the future. It only takes a minute. Thanks!